Manometer for pressure pmax = 60 mbar • Ficha de Dados
Suitable for: RITTER Bellows-type Gas Meters pmax = 60 mbar
|Type||Capsule pressure gauge|
|Measuring Range||0 to 60 mbar|
Material: Stainless steel (Cr-Ni)
The Manometer for pressure pmax = 60 mbar can be used for measurement of the gas pressure while measuring the gas flow. Among other reasons, this is necessary if the measured and indicated actual volume of gas must be recalculated into the norm volume. The actual volume is the volume at the actual temperature and the actual pressure.
The norm volume of a gas is the volume at norm conditions which are (in Germany):
|Norm Temperatur||= 273,15 Kelvin ( = 0 °C)|
|Norm Druck||= 1.013,25 mbar|
The formula for converting the actual volume into norm volume is:
Note: The indicated gas pressure at the manometer is the differential pressure between the gas pressure at the gas inlet and the actual atmospheric air pressure. Thus, the actual gas pressure (pa) to use in the above formula is the indicated gas pressure at the manometer plus the actual atmospheric air pressure in (mbar).
Unpack the Manometer which is mounted into a T-piece. According to the rules for calibration and measurement with gas meters, the Manometer must be positioned at the gas inlet of the meter. The gas inlet nozzle is labelled accordingly.
Mount the Manometer onto the gas inlet nozzle by tightly screwing the union nut which is attached to the Thermometer. Thus, the Manometer is ready for use.
V 2.3 / Rev. 2022-04-05 / Sujeito a alterações.
Dr.-Ing. RITTER Apparatebau GmbH & Co. KG · Coloniastrasse 19-23 · D-44892 Bochum · Alemanha
Para perguntas, favor contatar firstname.lastname@example.org ou seu parceiro de distribuição local (em nossa página de resumo)