Generatore di impulsi V3.2-50 »Smart PG« · Scheda dati
- 1 Quick reference
- 2 Application
- 3 Components
- 4 Description
- 5 Output Socket
- 6 Sample circuit
- 7 Use with Drum-type Gas Meters
- 8 Performance Data
- 9 Temperature range
- 10 Humidity range
- 11 Electrical Data
- 12 Specifications
- 13 Connection of the Pulse Generator to the Electronic Display Unit »EDU 32 FP« (optional accessory)
- 14 Dimensions of encoding disc
- 15 Exchanging the spare parts kit »Complete Photo diode«
The Pulse Generator for RITTER Drum-type Gas Meters and Bellows-type Gas Meters is a rotary encoder for pulse output. It can be used to transfer the measured gas volume for remote display and/or data processing (calculation of flow rate, data transfer via RS232) to the Electronic Display Unit »EDU 32 FP« (accessory) or to an external measuring system / PC. In the latter case, the external system must provide the power supply (5 – 24 V) for the photo sensor as well as the evaluation circuit/logic which enables the direct readout of the measured volume and flow rate. For connection to an external system, please refer to »Electrical Data« and the wiring diagrams.
The Pulse Generator is located within the housing of the counter mechanism of the Gas Meter (behind the dial plate) and it consists of the following components:
The measuring drum of drum-type meters and the measuring unit of bellow-type meters are coupled 1:1 to the slit disc via a magnetic coupling. The slits/flags of the slit disc rotate through the U-shaped photo sensor, thereby interrupting the light beam of the photo diode intermittently. Thus, the photo interrupter converts the revolution of the measuring drum into a sequence of pulses. The number of pulses represents the volume of gas which has passed through the Gas Meter, depending on the respective resolution (see »Performance Data«). The frequency of the sequence of pulses is a measure of the rotational speed of the measuring drum and thereby a measure of the flow rate of the gas.
For operation of the photo sensor, an external electric power supply in the range of 5 – 24 Volts DC is required. More electrical data are stated under »Electrical Data«. The output signal is a TTL signal, whereby the pulse level (= min./max. voltage of the signal) depends on the power supply:
For power supply values between 5 and 24 Volts, the output signal level can be linearly interpolated for the first approximation.
The pin connection of the 5-pin output socket is shown under »Pin configuration of the Output Socket«.
The connection of a measurement instrument to the Pulse Generator is shown schematically under »Wiring diagram / sample circuit«.
Use with Drum-type Gas Meters
Drum-type gas meters are volumetric gas meters. That means, that they are measuring gas volume precisely. When the Pulse Generator is used with drum-type gas meters for recording the gas flow, it is possible for the respective Voltage Output curve (line) to be wavy, even when gas flow is constant. This is (unpreventably) caused by the type of construction of the measuring drum: the drum consists of four separate chambers, which are closed and opened in sequence. The previous chamber has to be closed before the next chamber will open.
This compulsory measurement is the reason for the high measurement accuracy. However, each closing also causes a little build-up of pressure at the inside of a chamber. The surface tension creates an additional pressure increase during emerging of a chamber (water highest surface tension, oil: lower, CalRix lowest). The resulting pressure increase causes a small reduction in the rotational speed of the measuring drum. This is barely visible to the eye but is documented precisely by a computer/transcriber. Thus, the wavy output line at constant input flow documents the true flow through the gas meter.
|Pulses per Revolution*|
|Gas Flow per Revolution*|
|Pulses per Liter|
|Maximum Pulse Frequency|
|TG 01||not applicable|
* TG types: Revolution of measuring drum ( = revolution of large needle of dial plate) / BG types: Revolution of large needle of dial plate
At higher temperatures the Pulse Generator must be cooled by flushing the counter mechanism casing with room air.
Necessary equipment: Optional connection nozzle at counter mechanism casing.
With a higher humidity, the circuit board of the Pulse Generator can be covered with a protective lacquer. Please indicate prior to order.
|Supply voltage Us||5 – 24 V DC|
|Supply current||< 3 mA|
|Voltage output Us = 5 V||high level||min. 3.7 V|
|low level||max. 0.7 V|
|Voltage output Us = 24 V||high level||min. 21 V|
|low level||max. 2 V|
|Current output||Source||min. +7 mA|
|Sink||min. -6 mA|
|Operating frequency photo diode||0 – 250 Hz|
Pin configuration of the Output Socket
Wiring diagram / sample circuit (schematic)
Connection of the Pulse Generator to the Electronic Display Unit »EDU 32 FP« (optional accessory)
The Pulse Generator can be connected to the optional accessory »Electronic Display Unit« by means of the 3-pin connection cord, which is supplied in conjunction with the Electronic Display Unit. The Electronic Display Unit contains the power supply for the inductive sensor as well as the evaluation circuit/logic which enables the direct readout of the measured volume (ltr) and flow rate (ltr/h).
Wiring of the Pulse Generator to the EDU socket
The measurement results displayed by the Electronic Display Unit can be transmitted to a computer via the standard-type interface RS 232 (please refer to the EDU Operation Instructions, paragraph 7.3 as well). Additionally, the value of the flow rate can be transmitted to an analog measurement device via the standard-type analog output (0 – 1 Volt or 4 – 20 mA).
Set-up of EDU
Dimensions of encoding disc
|TG05 to TG50 / BG|
|Slit width||4.2 mm|
|Flag width||4.4 mm|
Exchanging the spare parts kit »Complete Photo diode«
The kit consists of the following components which are already mounted on a transparent plastic cover plate:
Removal of the built-in kit:
Replacement with the new kit:
V 3.2 / Rev. 2023-03-24 / Con riserva di modifiche.
Dr.-Ing. RITTER Apparatebau GmbH & Co. KG · Coloniastrasse 19-23 · D-44892 Bochum · Germania
Per domande si prega di contattare firstname.lastname@example.org o il tuo partner di distribuzione locale (nella nostra pagina panoramica)