Manometer for pressure pmax ≤ 1 bar • डाटा शीट
Suitable for: RITTER Drum-type Gas Meters 500 mbar ≤ pmax ≤ 1 bar
|Type||Capsule pressure gauge|
|Measuring Range||0 to 600 mbar||0 to 1 bar|
|Resolution||20 mbar||50 mbar|
Material: Stainless steel (Cr-Ni)
The Manometer can be used for measurement of the gas pressure while measuring the gas flow. Among other reasons, this is necessary if the measured and indicated actual volume of gas must be recalculated into the norm volume. The actual volume is the volume at the actual temperature and the actual pressure.
The norm volume of a gas is the volume at norm conditions which are (in Germany):
|Norm temperature||= 273,15 Kelvin ( = 0 °C)|
|Norm pressure||= 1.013,25 mbar|
The formula for converting the actual volume into norm volume is:
|VN||=||Norm Volume in||(ltr)|
|Vi||=||indicated Volumen in||(ltr)|
|pN||=||Norm Pressure in||(mbar-absolut)|
|pa||=||actual Pressure in||(mbar-absolut)|
|TN||=||Norm Temperature in||(Kelvin)|
|Ti||=||indicated Temperature in||(Kelvin)|
Note: The indicated gas pressure at the manometer is the differential pressure between the gas pressure at the gas inlet and the actual atmospheric air pressure. Thus, the actual gas pressure (pa) of the above formula equals the indicated gas pressure at the Manometer plus the actual atmospheric air pressure in (mbar).
The Manometer is pre-mounted to the gas meter (positioned at the gas inlet nozzle). The gas inlet nozzle is labelled accordingly. Therefore, the manometer is ready for use and no further installation is to be performed by the user.
Please note: The manometer screw connection to the gas meter is sealed by Teflon® tape. When disassembling the manometer from the gas meter, the Teflon® tape cannot be used again and must be replaced by a new Teflon® tape.
Image capsule pressure gauge
Source: WIKA Alexander Wiegand SE & Co. KG, 63911 Klingenberg/Germany, Data sheet PM 06.03 . 04/2017
V 1.0 / Rev. 2022-06-27 / Subject to alterations.
Dr.-Ing. RITTER Apparatebau GmbH & Co. KG · Coloniastrasse 19-23 · D-44892 Bochum · Germany
For questions please contact firstname.lastname@example.org or your local distribution partner (on our overview page)