Thermometer Gas (TG) • Data Sheet
Suitable for: RITTER Drum-type Gas Meters
Filling liquid: Isoamylbenzoat »IAB«
|0 °C||to + 60 °C,||scale graduation 0.5 °C|
|0 °C||to + 80 °C||scale graduation 0.5 °C|
|15 °C||to + 30 °C||scale graduation 0.1 °C|
|0 °C||to + 50 °C||scale graduation 0.1 °C|
The Thermometer can be used for measurement of the gas temperature while measuring the gas flow. Among other reasons, this is necessary if the measured and indicated actual volume of gas must be recalculated into the norm volume. The actual volume is the volume at the actual temperature and the actual pressure. The norm volume of a gas is the volume at norm conditions which are (in Germany).
The norm volume of a gas is the volume at norm conditions which are (in Germany):
|Norm Temperatur||= 273,15 Kelvin ( = 0 °C)|
|Norm Druck||= 1.013,25 mbar|
The formula for converting the actual volume into norm volume is:
Unpack the Thermometer. Unscrew the closing cap of the Thermometer (Gas) support on the Gas Meter. Mount the Thermometer by inserting it carefully through the Thermometer (Gas) support. Seal the Gas Meter's casing by tightly screwing the union nut which is attached to the Thermometer. Thus, the Thermometer is ready for use. The removed closing cap of the support can be stored easily by screwing it onto the respective thread support located at the base plate of the gas meter.
V 2.0 / Rev. 2020-01-30 / Subject to alterations.
The most recent version of this data-sheet can be found at https://www.ritter.de/en/data-sheets/thermometer-tg-gas/
Dr.-Ing. RITTER Apparatebau GmbH & Co. KG · Coloniastrasse 19-23 · D-44892 Bochum · Germany
For questions please contact email@example.com or your any local distributor at https://www.ritter.de/en/worldwide/